3.keep the chamber cover clean,and press the cover tightly to let it suck the chamber
1.There is something leak, according to the instructions to determine the leak point (pump, pipe joints, barrels), if it is barrel leak, and then check which specific parts of the leak (table, valve, silicone ring), we will send you the relevant accessories.
2. Silicone rings may be dusty and please clean or wipe it with water.
3. When you vacuum, please press the silicone ring
4. Uneven edge of vacuum chamber will lead to leakage (the possibility is relatively small because of full inspection), if it is, we will resend a new barrel to you.
Q2. Were you able to pull a full 29 inHg of vacuum? I could only get 27 inHg.
Normal vacuum pressure gauges show relative pressure. Therefore any measurements are dependent on the atmospheric pressure, which is dependent on height above sea level and air conditions. Let’s say we have a certain vacuum at sea level and the gauge shows -0.095 MPa(-28 inHg). At 900 m or 2950 ft height above sea level the same vacuum gives -0.085 MPa (-25in Hg)on the gauge, although the vacuum quality in the two situations is exactly the same. If you ensure no leaks, -29~-27 inHg is regarded as normal.
Q3. I need the vacuum of -29 inHg, why does the pressure gauge show only max -14?
0.1 MPa=100,000 Pa=14.5 PSI=29.5 inHg
As you see, our vacuum gauge measuring range is 0~-0.1 MPa/-14 PSI
If you keep every connectors sealed and operate correctly, the negative pressure can reach at least -14 PSI(-28.5 inHg) even -0.1 MPa(-29.5 inHg). If you cannot get the result, that means there is a leak. Please read our instruction and then check leak.
Q4. Parts & Fittings for Vacuum Series
Please visit our website to purchase:
Q5. When I brought the chamber down to 12psi vacuum cracks radiated from both the vacuum and gauge holes drilled into the acrylic lid. What can be done about the problem?
a. According to our experience, some applications may cause lid crack such as stabilizing wood with cactus juice. Please do not evacuate corrosive chemical materials.
b. Continuing to pull vacuum after the needle has stopped moving may cause crack.
c. Please don't maintain chamber in a high vacuum(or full vacuum) for more than 24 hours.
d. In a high-vacuum state, please do not collide and hit the vacuum chamber, also try not to move to avoid collisions.
e. Acrylic is fragile, please avoid falling from height.
Please purchase glass lid vacuum chamber:https://bacoeng.com/collections/main-chamber-pump-accessories/products/glass-lid-vacuum-chamber
Q6. Oil leaked from vacuum chamber pressure gauge. Is the vacuum chamber still work? Do I need to add oil?
A: The main function of oil is to prevent gauge from damage in the shaking environment during transit. As the small rubber bit of the pressure gauge not covered, the oil was leaked during the transit. So if you use it normally(Not used in tremulous conditions), oil doesn't work for you.
Please clean up the oil. If you need to fill gauge with oil, We suggest you using silicone oil. Some people use glycerol, but the silicone oil is more resistant to low temperature, and its stability is better.
Q7. I have the BacoEng 1.5 gallon stainless vacuum chamber with the clear lid. I use it to degas two part urethanes. Why the silicone sealing ring is expanding?
A: We presume it is absorbing some of the component gasses. Silica gel rings are consumable and can be replaced. Please purchase it on our official website.
Q8. What's the square silicone pad for?
A：Food grade silicone pads. If the material you put inside the chamber is solid, you can put it directly on the silicone pad. Or it can be used to prevent contamination.